Home » Engineering » RotaMeter Principle, Working, Construction, and Types

# RotaMeter Principle, Working, Construction, and Types

Rotameter is a safe, easy, and reasonably priced flow measurement tool. The flow rate of a liquid or a gas is thus measured with this equipment. This meter has a tapered tube that passes through a floating internal component. Rotameters are also known by the names gravity flow meters, mechanical flow meters, or variable area flow meters. The term “gravity” refers to the requirement that the rotameter is installed vertically in gravity flow meters.

## Rotameter Principle

It consists of a verticle, a tapered and transparent tube in which a plummet is placed. During the fluid flow through the tube, the plummet rises and falls because of variation in flow. The Rotameter principle works on of force of gravity and upthrust force exerted by a fluid.

## Rota Meter Construction

• It consists of a verticle tapered tube, which is mounted with a narrow end down.
• The tube is usually made of glass to which a linear scale is attached.
• A solid plummet is placed in the tube (the diameter is smaller than the narrow end of the tube).

## Working:

The first rotameter is fitted on a pipe, as flow is moving in an upward direction through a tapered tube, and the flow of fluid varies. the plummet, which is surrounded by the fluid, rises and falls depending on the rate of flow, the greater the flow rate, the higher the plummet rise in the tube.

Due to gravity and drag, respectively, the float in the rotameter is subject to two forces acting in opposite directions.

The float is pulled upward from gravity by the liquid flow. After a while, the flowing area reaches a place where the force applied to the floating body is exactly equal to the weight of the plummet. Therefore, the plummet will reach equilibrium once the area where it is moving causes enough drag to equal its own weight.

The distance the plummet is raised when gravity and its weight are both stable depends on how quickly the liquid is moving through the tapered tube.

The flow may be read using the upper edge of the plummet as an index (linear scale). The reading can be sent for recording, integrating, and controlling purposes.

## Types of Rotameters

Basically, Rotameter is three types:

1. Glass tube rota meter
2. Metal tube rotameter
3. Heavy duty rotameter

### 1. Glass tube rotameter

As the name indicates, it s made of borosilicate glass, and float can be manufactured by metal (usually stainless steel ), glass, and plastic.

Glass tube rotameter can be used in water over 90°C which is high pH that softens glass: wet steam.

Glass tube rotameter cannot use in caustic soda and hydrofluoric acid.

### 2. Metal tube rotameter

Made of aluminum, brass, or stainless steel, so it is easy to use for higher temperatures and pressure.

Since they are not transparent, mechanical or magnetic flow the outside of the tube is used to determine the float position.

The spring and piston combination determine the flow rate, it is easy o used with fluids like hot and strong alkalis, fluorine, hydrofluoric acid, hot water, steam, sour gas, additives, and molten metals.

They can be operated up to 750 psi, temperature to 540°C, and can measure flow up to 4000 GPM of water or 1300 scfm of air.

### 3. Heavy Duty/ Industrial pressure transducers:

Heavy Duty/ Industrial pressure transducers feature a much more rugged enclosure than other transduce.

They are intended for use in tough industrial environments.

They also frequently have a scaled 4-20mA output, which gives significantly higher resistance to electrical noise, which is prevalent in industrial settings.

Read Also: What is Pitot Tube

#### Parameters

The following parameters need to consider while selecting rotameters.

• Operating pressure
• Temperature of liquid
• Diameter of pipe
• Indicating the type of scale
• Requirements of valve
##### Application of the Rotameter
• It is used in the chemical industries, and oil industries.
• It is used for gas and Liquid
• Used to measure oxygen flow rate in medical areas.

• Rota meter required no external power or fuel, it used only the inherited properties of the fluids along with gravity to measure the flow rate.
• A Rotameter is also a relatively simple device that can be mass-manufactured out of cheap materials, allowing it for widespread usage.
• Clean glass is used which is highly resistant to thermal shock and chemical reactions.